for standard waveforms. The Agilent Technologies A Use the A to generate complex . Manual language options (please specify one). ABA US. For Safety information, Warranties, and Regulatory information, see the last page in this manual. Agilent A. Option Phase-Lock Assembly. oscilloscope or DMM and send it to your HP A for output. 3-year warranty. With your HP A, you get operating and service manuals, a quick reference .
|Published (Last):||15 June 2018|
|PDF File Size:||8.48 Mb|
|ePub File Size:||18.43 Mb|
|Price:||Free* [*Free Regsitration Required]|
Numeric Parameters Commands that require numeric manuak will accept all commonly used decimal representations of numbers including optional signs, decimal points, and scientific notation. SineSquare, Triangle, Ramp, or Arbitrary waveform. The Surplus Equipment Pros. Freq 1 Enable the frequency modify mode. The function generator calculates the average of the data points and compares the average to zero volts.
Internal-External both or External only. See terms – opens in a new window or manuql. The function generator is configured manua, 8 data bits with no parity when shipped from the factory. Page Index front panel annunciators, 5 factory setting gate source dc volts selection, 24 baud rate RS, burst modulation, downloading waveforms, – calibration security code, gated burst mode, 81, menu examples, 33 – 39 data bits RS, GPIB menu messages, The carrier output can be controlled using either triggered or externally- gated methods.
Special financing available Select PayPal Credit at checkout to have the option to pay over time. Learn more – opens in a new window or tab. This section is only an SCPI users, see page The phase accumulator output the upper 14 bits will step sequentially Chapter 3 Features and Functions Calibration Overview Calibration Overview This section gives an overview of the calibration features of the function generator.
The following steps show you how to enter a number in the menu. Therefore, the value is the most useful way to specify ac signal amplitudes.
Be the first to write a review. In this example, you will enable the function generator to automatically recall the power-down state when power is turned on. I can go for that if i’m sure that it does not cause any other effect. You may have misspelled the command or it may not be a valid command. Government prime and Service Office. Chapter 7 Tutorial Creating Arbitrary Waveforms Leakage error is caused when the waveform record does not include an integer number of cycles of the fundamental frequency.
Page 5 Front-Panel Number Entry You can enter numbers from the front-panel using one of three methods. For additional information, see the Global Shipping Program terms and conditions – opens in a new window or tab No additional import charges on delivery Delivery: Bao Ho Bao Ho view profile. Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab.
The default is INT. Table Of Contents Contents Chapter 1 Quick Start To prepare the function generator for use 15 If the function generator does not turn on 16 To adjust the carrying handle 18 To set the output frequency 19 To set the output amplitude 20 To set a dc offset voltage 21 To set the duty cycle 22 To output a stored arbitrary waveform Output Minimum Maximum Function Termination Seller does not offer returns.
The terminated output amplitude is calibrated for an exact 50 ohm load.
Query the number of non-volatile memory slots available to store user-defined waveforms. OFFS commands to select the function, frequency, amplitude, and offset of the carrier waveform. The following steps show you how to change the frequency to 1. To 333120a proper operation, you should enable the sweep mode after you have set up the other sweep parameters. To recall the stored state, you must use the same memory location used previously to store the state.
For certain applications where you may not want to output the sync signal, maanual can disable the SYNC terminal. PNP transistor not working 2.
Page 19 Chapter 1 Quick Start If the function generator does not turn on Remove the line-voltage selector from. Updates 1 Dec 1. All of the standard output functions except dc and noise have an associated sync signal. Chapter 3 Features and Functions Output Configuration Output Configuration This section contains information to help you configure the function generator for outputting waveforms.
Chapter 3 Features and Functions Amplitude Modulation AM Modulating Waveform Shape The function generator will accept an internal modulating signal, an external modulating signal, or both. Set the carrier frequency between 10 mHz and 15 MHz kHz for triangle and ramp. Michael Grieger 331200a Jul Please enter 5 or 9 numbers for the ZIP Code.
Chapter 1 Quick Start To store the instrument state To verify that the state was stored properly, you can turn the power off before recalling the state. Throughout this manual, the following conventions are used for command syntax. See the next page if you need to change the power-line fuse.
ModelSim – How to force a struct type written in SystemVerilog?
Chapter 2 Front-Panel Menu Operation A front-panel menu tutorial Menu Example 1 The following steps show you how to turn on the menu, move up and down between levels, move across the choices on each level, and turn off the menu.
Chapter 3 Features and Functions Calibration Overview To Secure Against Calibration You can secure the majual generator against accidental calibration either from the front panel or over the remote interface. Based on your location, we recommend that you select: