Gonotec – Osmomat The OSMOMAT is the standard system for the determination of the total osmolality in aqueous solutions. This system is commonly. Gonotec Gesellschaft für Meß- und Regeltechnik mbH does not assert any claim on trademarks or trade names other than its own. ® OSMOMAT model. Find pricing on new and used Gonotec Osmomat ‘s and other lab products. Connect with a variety of sellers on LabX and buy today.
|Published (Last):||23 October 2008|
|PDF File Size:||15.93 Mb|
|ePub File Size:||3.22 Mb|
|Price:||Free* [*Free Regsitration Required]|
In the idle position of the motor swipe, the end of the needle must not extend below the lower edge of the cooling pin.
Os,omat measuring vessels also exhibit these characteristics. Inadequate servicing or improper modification; e. It is thus possible to set the osmolality value of the calibration solution to be used.
Provisions of Certification CE compliance requires that the unit be installed and operated in the manner described in this manual. The unit must be adequately disinfected prior to decommissioning. Approximately 7 kg Option 3-P: The iced-covered tip of the stainless steel needle passes through a bore above the measuring vessel during the crystallization-initiating phase.
Its purpose is to provide information to help identify, diagnose, monitor and treat physiological conditions, states of health or illnesses. This is because their openings are dilated when they are placed on the sensor.
Ensure that you have the equipment documentation handy. It should engage slightly. Seed crystals of this type can be easily eliminated by briefly heating the sample liquid close to its boiling point osmlmat subsequently cooling it. Contracts for additional warranty services, such as on-site service, may be available from an authorized Gonotec sales partner.
If the air humidity is very low for example, in heated or airconditioned roomsit may be necessary to open the cover of the upper cooling system. The ksmomat may have other legal claims that go beyond those outlined here. The only remedy in such cases is to adequately dilute the sample solution. When the measuring vessel is placed on the thermistor probe, a soft clicking sound is heard and there is a tangible resistance.
The crystallization enthalpy causes an immediate rise in temperature up to the freezing point of the remaining, somewhat more highly concentrated solution. We have been able to automate the draining of the measuring system so that it is possible to reliably obtain a high sample throughput.
Any departure from the specifications or independent modifications of the calibration- and reference material without the express consent of Gonotec GmbH may result in a violation of CE requirements.
It is simple to operate, very reliable and provides extremely rapid sample handling.
No crystallization Possible Cause Procedure Needle too long or too short Bring needle to proper length Needle not deburred at tip e. No crystallization Bar of needle disengaged from motor or stuck firmly in place Check upper cooling system component group Fuses in power switch blow out when unit is switched on The following parts may be damaged: A repeat determination using just one sample is not possible, as the sample separates during measurement.
The purpose of the adjustment is align the center axis of the cooling clamp with the thermistor axis see Figure 7: Thermistor probe adjustment along X-axisthen carefully tighten osmmat retaining screws.
Please note the following important restrictions and operational guidelines: The measuring vessel holder is lowered and the cooling process begins. Respective Jurisdiction This warranty statement guarantees the customer certain legal claims.
Clean the sensor thoroughly using water and a soft paper towel to remove any seed crystals. The simple menu-driven operation requires a minimum of user interaction and allows an automated operation of the Osmomat. If the power cord plug does not match the conventional power outlets in your location, you may substitute another power cord.
The calibration- and reference material is a chemical. Extension of the Limited Warranty Gonotec warrants the end user that Gonotec products shall be free from manufacturing and material defects for the above periods of validity from the purchase date.